Feasibility of aortic valve assessment with low dose prospectively triggered adaptive systolic (PTAS) cardiac computed tomography angiography
© Lee et al.; licensee BioMed Central Ltd. 2013
Received: 25 January 2013
Accepted: 12 April 2013
Published: 20 April 2013
Cardiac computed tomography angiography (CTA) is feasible for aortic valve evaluation, but retrospective gated protocols required high radiation doses for aortic valve assessment. A prospectively triggered adaptive systolic (PTAS) cardiac CT protocol was recently described in arrhythmia using second-generation dual-source CT. In this study, we sought to evaluate the feasibility of PTAS CTA to assess the aortic valve at a low radiation dose.
A retrospective cohort of 29 consecutive patients whom underwent PTAS protocols for clinical indications other than aortic valve assessment and whom also received echocardiography within 2 months of CT, was identified. Images were reviewed for aortic valve morphology (tricuspid/bicuspid/prosthetic) and stenosis (AS) by experienced blinded readers. Accuracy versus echocardiography and radiation doses were assessed.
All PTAS coronary CTAs were clinically diagnostic with 0 un-evaluable coronary segments. The accuracy of PTAS for aortic valve morphology was 92.6%, and for exclusion of severe AS was 93.1%. Two exams were un-evaluable for the aortic valve due to inadequate number of phases archived for interpretation. Total radiation dose was a median of 2.8 mSv (interquartile range 1.4–4.4 mSv).
PTAS CTA protocols using second-generation dual-source CT for aortic valve evaluation are feasible at low doses. This protocol should be investigated further in larger cohorts.
KeywordsComputed tomography angiography Aortic stenosis Low dose
Prospectively ECG-triggered cardiac CT angiography targeted to end-systole for the evaluation of aortic pathology was feasible at a low radiation dose (2.8 mSv) with 128 dual-source CT.
The feasibility of cardiac computed tomography angiography (CTA) for aortic valvular evaluation has been established for aortic stenosis (AS) evaluation (via direct planimetry) in small referral cohorts [1–9], and has performed well versus transthoracic echocardiography. All initially published studies were performed using retrospective ECG gating to allow systolic phase evaluation of aortic valve opening [10, 11]. Retrospective ECG gating is widely available but involves significant radiation dose expense versus prospective triggering and is now used infrequently [12, 13].
Various refinements to prospective triggering algorithms such as arrhythmia rejection are available [13–15]. Recently, the use of a systolic target for prospective triggering was described as a means to mitigate variable and elevated heart rates [16–18]; this method was applied using late-generation dual-source CT scanners, which allowed a “systolic absolute delay” to specify an acquisition in systole after the R-peak, and a fixed range of data acquisition to enable various reconstruction intervals to achieve motion-free images. This method limits imaging to mid and late systole, phases shown to be ideal for aortic valve area opening measurements .
Because this protocol may allow aortic valve analysis (valve morphology and AS), we evaluated its feasibility versus echocardiography. We hypothesized that prospectively triggered adaptive systolic images (PTAS) would allow adequate aortic valve evaluation at a significantly decreased radiation dose versus retrospectively ECG-gated images (6.7 to 20 mSv in one meta-analysis) .
Our institutional IRB (The Partners Human Research Committees) granted a waiver for this retrospective research. HIPAA compliance was maintained throughout the study. No outside funding was received, and the authors maintained full control over the data.
This retrospective study included 29 patients who underwent clinically indicated CTA with PTAS from December 2011 to August 2012. All patients underwent echocardiography within two months of CTA.
Image reconstruction and analysis
Raw data were reconstructed at multiple phases of R-R interval (300 msec, 350 msec, 375 msec, 400 msec, and/or as clinically necessary). For each phase archived in PACS, images were reconstructed in multiplanar reformatted axes to obtain a series of short axis view through the aortic valve.
Volume CT dose index (CTDIvol) and dose length product (DLP) for the CTA scans were recorded and effective dose was calculated .
Aortic valve evaluation
Two readers each with 4 years each of advanced training (2 years of echocardiography and 2 additional years of cardiac CT fellowships) were blinded to CTA and echocardiography results. Readers reviewed images in consensus and assessed aortic valve morphology and AS. The aortic valve was assessed as 1) tricuspid, 2) bicuspid, or 3) prosthetic. Confidence in their assessment was graded on a 4-point Likert scale (1 = not at all, 2 = somewhat, 3 = confident, 4 = very confident). Aortic valve orifice area was measured for each available phase [2, 11]. The maximum area was used to grade potential AS (no evidence of severe AS vs. severe AS [area less than 1 cm2 by planimetry]) [2, 4]. If severe AS could be excluded, confidence was graded in a similar 4-point Likert scale. Clinical echocardiography reports were independently reviewed by a research fellow. In echocardiography, aortic valve area was evaluated by the continuity equation with valve area less than 1 cm2 considered severe stenosis. Agreement tables between PTAS CTA and echocardiography were compared.
Parametric variables were expressed in mean ± standard deviation. Non-parametric variables were expressed in median and interquartile range (IQR). Categorical variables were expressed as percentages. The accuracy of PTAS CTA was calculated by (number of cases with result agreement between CTA and echocardiography) / (total number of cases evaluable by echocardiography).
Patients characteristics and scan parameters n = 29
64.1 ± 16.6
Mean heart rate (bpm)
66.4 ± 10.9
Heart rate variability (bpm)
39.9 ± 35.6
Rhythm during scan
Other (atrial flutter, etc.)
Contrast amount (cc)
96.1 ± 13.9
Flow rate (cc/sec)
6.0 ± 2.8
Beta blocker used
24.9 ± 4.6
PTAS CTA radiation exposure by tube potential and scan indication
Tube current (mAs)
Scan length (cm)
Effective Radiation (mSv)
218.0 ± 40.2
Radiation exposure by tube voltage
80 kV (n = 8)
211.4 ± 42.0
100 kV (n = 10)
202.8 ± 36.8
120 kV (n = 7)
222.1 ± 35.3
140 kV (n = 4)
262.0 ± 29.7
Radiation exposure by scan indication
Native coronary (n = 19)
225.7 ± 40.1
Bypass graft (n = 3)
232.0 ± 17.1
Aorta (n = 5)
183.2 ± 35.1
Pre-TAVI (n = 2)
211.0 ± 53.7
20.7 [ 20.7–20.7]
Valve morphology agreement table between PTAS CTA and echocardiography
Not assessed/ not evaluated
AS agreement table between PTAS CTA and echocardiography
No severe aortic stenosis
Severe aortic stenosis
No severe aortic stenosis
Severe aortic stenosis
Overall performance of PTAS CTA on aortic valve evaluation
3.2 ± 1.1
Severe aortic stenosis
3.4 ± 1.1
Reasons for low confidence were similar for aortic valve morphology and AS assessment, such as that the aortic valve remained in closed position throughout all phases (n = 2 for both), motion artifact (n = 1 for valve morphology, n = 2 for AS assessment), high image noise (n = 1 for both), and poor contrast opacification (n = 1 for both).
We evaluated the feasibility of a novel protocol to allow prospectively ECG-triggered cardiac CT angiography targeted to systolic phases for secondary evaluation of aortic valvular pathology. We demonstrated that when adequate phases of the R-R interval were available, the aortic valve could be evaluated with low radiation doses (2.8 [1.4–4.4] mSv). An important caveat in this analysis is that patients were referred primarily for coronary arterial or bypass graft (n = 21) and aortic root or annulus (n = 5) evaluation (with fully diagnostic results for all primary indications).
PTAS CTA was initially described as a means of limiting radiation dose in the setting of heart rate irregularity [16–18]. Previously, Feuchtner et al demonstrated the feasibility of assessing valvular function and morphology in patients with stable sinus rhythm. However, we note that in that study, only patients with high heart rate (>65 bpm) received systolic triggered exams with widened window acquisition (patients with low heart rate underwent diastolic triggered exams). In addition, those CTAs were performed with additional 20% mAs exposure during 10–90% of the cardiac cycle. In our present study, the widened acquisition window with 20% mAs exposure was only used from 200 to 450 msec after the R peak, or approximately 20%–50% of the cardiac cycle for patients with a heart rate of 66 bpm (the average heart rate in our cohort). Therefore, while Feuchtner et al showed accurate aortic assessment with 20% mAs exposure during nearly the entire cardiac cycle, we demonstrated that systolic phase image acquisition with a 250 msec acquisition window may be sufficient for aortic assessment.
The two most effective methods of cardiac CT dose reduction are the use of prospective triggering and tube potential lowering [13, 14, 22]. Traditionally, prospectively triggered CTAs were prone to motion or misregistration artifacts caused by irregular heart rhythms. In fact, atrial fibrillation was formerly a relative contraindication to CTA due to high radiation exposure and image quality concerns . This challenge posed by elevated and irregular heart rates and rhythms has been progressively addressed by technologic developments such as arrhythmia rejection algorithms  and systolic triggering .
In addition to prospective ECG triggering, we implemented automatic kVp selection to minimize tube voltage and concomitantly adjust tube current to preserve the necessary photon flux [20, 21]. Some cases involved higher kV settings due to large patients and longer scans of the entire chest since the scans were not clinically performed for aortic valve assessment.
Concurrent assessment of aortic valve by PTAS CTA in addition to coronary assessment carries several clinical advantages. If aortic valve pathology is unknown or unexpected, obvious benefits can be realized through early diagnosis. In cases of thoracic aortic evaluation, the questions of aortic valve morphology (such as bicuspid versus tricuspid) might be addressed, thereby obviating transesophageal echocardiography in some cases. In the setting of known aortic valvular disease (such as in percutaneous aortic valve implantation planning CTA), a PTAS protocol can allow concurrent valvular assessment and coronary artery stenosis exclusion. This could obviate the need for preoperative invasive coronary angiography (acknowledging that these are older, higher risk patients than traditional CTA patients), and potentially obviate the need for pre-procedural transesophageal echocardiography. While the traditional assessment of the significance of aortic stenosis is by Doppler evaluation, direct planimetry is performed by transesophageal echocardiography; this can be assessed by CTA .
Our retrospective cohort study carries several limitations precisely because patients were referred for other indications than primary aortic valve assessment. While images were sufficient for the clinically requested coronary or aortic vascular assessment in all cases, insufficient aortic valvular phase images were archived in 2 cases. In addition, although adequate imaging may have been initially acquired for aortic valve analysis, not all necessary phases were available for interpretation in our study (i.e. were not archived). Second, we utilized a second-generation dual-source scanner, with the most current software upgrade. Other scanners have been used for aortic valve imaging, but may not have available similar settings such as prospective triggering options with positive absolute phase start times or high temporal resolution . Other equipment therefore may require different triggering settings or require longer acquisitions and larger radiation doses. Our method therefore warrants dedicated confirmatory study in a larger cohort on this specific scanner, and exploratory study using other scanner models which offer prospective ECG triggering.
We demonstrated feasibility of a prospectively ECG-triggered cardiac CT angiography targeted to end-systole for the evaluation of aortic pathology at a low radiation dose (2.8 mSv) with echocardiography as the reference standard. This is significantly lower than previously reported retrospectively ECG-gated modes (6.7 to 20 mSv). We note that care must be taken to reconstruct sufficient phase reconstructions to completely evaluate the aortic valve. Our results suggest that further study in a larger cohort should be performed to confirm our favorable results and to ensure the maintained diagnostic capacity of native coronary arteries.
Our institutional IRB (The Partners Human Research Committees) granted a waiver for this retrospective research. Patient consent was not necessary due to the retrospective nature of this study.
Availability of supporting data
Data available upon request.
- Alkadhi H, Wildermuth S, Plass A, Bettex D, Baumert B, Leschka S, Desbiolles LM, Marincek B, Boehm T: Aortic stenosis: comparative evaluation of 16-detector row CT and echocardiography. Radiology. 2006, 240: 47-55. 10.1148/radiol.2393050458.PubMedView ArticleGoogle Scholar
- Feuchtner GM, Muller S, Bonatti J, Schachner T, Velik-Salchner C, Pachinger O, Dichtl W: Sixty-four slice CT evaluation of aortic stenosis using planimetry of the aortic valve area. Am J Roentgenol. 2007, 189: 197-203. 10.2214/AJR.07.2069.View ArticleGoogle Scholar
- Chen JJ-S, Jeudy J, Thorn EM, White CS: Computed tomography assessment of valvular morphology, function, and disease. J Cardiovasc Comput Tomogr. 2009, 3: S47-S56. 10.1016/j.jcct.2008.10.017.PubMedView ArticleGoogle Scholar
- Chen JJ, Manning MA, Frazier AA, Jeudy J, White CS: CT angiography of the cardiac valves: normal, diseased, and postoperative appearances. RadioGraphics. 2009, 29: 1393-1412. 10.1148/rg.295095002.PubMedView ArticleGoogle Scholar
- Tanaka H, Shimada K, Yoshida K, Jissho S, Yoshikawa J, Yoshiyama M: The simultaneous assessment of aortic valve area and coronary artery stenosis using 16-slice multidetector-row computed tomography in patients with aortic stenosis comparison with echocardiography. Circ J. 2007, 71: 1593-1598. 10.1253/circj.71.1593.PubMedView ArticleGoogle Scholar
- Bouvier E, Logeart D, Sablayrolles JL, Feignoux J, Scheuble C, Touche T, Thabut G, Cohen-Solal A: Diagnosis of aortic valvular stenosis by multislice cardiac computed tomography. Eur Heart J. 2006, 27: 3033-3038. 10.1093/eurheartj/ehl273.PubMedView ArticleGoogle Scholar
- Feuchtner GM, Dichtl W, Friedrich GJ, Frick M, Alber H, Schachner T, Bonatti J, Mallouhi A, Frede T, Pachinger O: Multislice computed tomography for detection of patients with aortic valve stenosis and quantification of severity. J Am Coll Cardiol. 2006, 47: 1410-1417. 10.1016/j.jacc.2005.11.056.PubMedView ArticleGoogle Scholar
- Leborgne L, Choplin Y, Renard C, Claeys M, Levy F, Jarry G, Rey JL, Remond A, Quiret JC, Tribouilloy C: Quantification of aortic valve area with ECG-gated multi-detector spiral computed tomography in patients with aortic stenosis and comparison of two image analysis methods. Int J Cardiol. 2009, 135: 266-269. 10.1016/j.ijcard.2008.03.095.PubMedView ArticleGoogle Scholar
- Messika-Zeitoun D, Serfaty J-M, Laissy J-P, Berhili M, Brochet E, Iung B, Vahanian A: Assessment of the mitral valve area in patients with mitral stenosis by multislice computed tomography. J Am Coll Cardiol. 2006, 48: 411-413. 10.1016/j.jacc.2006.04.035.PubMedView ArticleGoogle Scholar
- Shah RG, Novaro GM, Blandon RJ, Whiteman MS, Asher CR, Kirsch J: Aortic valve area: meta-analysis of diagnostic performance of multi-detector computed tomography for aortic valve area measurements as compared to transthoracic echocardiography. Int J Cardiovasc Imaging. 2009, 25: 601-609. 10.1007/s10554-009-9464-z.PubMedView ArticleGoogle Scholar
- Abdulla J, Sivertsen J, Kofoed KF, Alkadhi H, LaBounty T, Abildstrom SZ, Kober L, Christensen E, Torp-Pedersen C: Evaluation of aortic valve stenosis by cardiac multislice computed tomography compared with echocardiography: a systematic review and meta-analysis. J Heart Valve Dis. 2009, 18: 634-643.PubMedGoogle Scholar
- Halliburton SS, Abbara S, Chen MY, Gentry R, Mahesh M, Raff GL, Shaw LJ, Hausleiter J: SCCT guidelines on radiation dose and dose-optimization strategies in cardiovascular CT. J Cardiovasc Comput Tomogr. 2011, 5: 198-224. 10.1016/j.jcct.2011.06.001.PubMedPubMed CentralView ArticleGoogle Scholar
- Ghoshhajra BB, Engel LC, Major GP, Goehler A, Techasith T, Verdini D, Do S, Liu B, Li X, Sala M: Evolution of coronary computed tomography radiation dose reduction at a tertiary referral center. Am J Med. 2012, 125: 764-772. 10.1016/j.amjmed.2011.10.036.PubMedView ArticleGoogle Scholar
- Hausleiter J, Meyer TS, Martuscelli E, Spagnolo P, Yamamoto H, Carrascosa P, Anger T, Lehmkuhl L, Alkadhi H, Martinoff S: Image quality and radiation exposure with prospectively ECG-triggered axial scanning for coronary CT angiography: the multicenter, multivendor, randomized PROTECTION-III study. JACC Cardiovasc Imaging. 2012, 5: 484-493. 10.1016/j.jcmg.2011.12.017.PubMedView ArticleGoogle Scholar
- Bittencourt MS, Schmidt B, Seltmann M, Muschiol G, Ropers D, Daniel WG, Achenbach S: Iterative reconstruction in image space (IRIS) in cardiac computed tomography: initial experience. Int J Cardiovasc Imaging. 2011, 27: 1081-1087. 10.1007/s10554-010-9756-3.PubMedView ArticleGoogle Scholar
- Lee AM, Engel LC, Shah B, Liew G, Sidhu MS, Kalra M, Abbara S, Brady TJ, Hoffmann U, Ghoshhajra BB: Coronary computed tomography angiography during arrhythmia: Radiation dose reduction with prospectively ECG-triggered axial and retrospectively ECG-gated helical 128-slice dual-source CT. J Cardiovasc Comput Tomogr. 2012, 6: 172-183 e172. 10.1016/j.jcct.2012.04.003.PubMedView ArticleGoogle Scholar
- Lee AM, Beaudoin J, Engel LC, Sidhu MS, Abbara S, Brady TJ, Hoffmann U, Ghoshhajra BB: Assessment of image quality and radiation dose of prospectively ECG-triggered adaptive dual-source coronary computed tomography angiography (cCTA) with arrhythmia rejection algorithm in systole versus diastole: a retrospective cohort study. Int J Cardiovasc Imaging. 2013, 24: 24-Google Scholar
- Feuchtner G, Goetti R, Plass A, Baumueller S, Stolzmann P, Scheffel H, Wieser M, Marincek B, Alkadhi H, Leschka S: Dual-step prospective ECG-triggered 128-slice dual-source CT for evaluation of coronary arteries and cardiac function without heart rate control: a technical note. Eur Radiol. 2010, 20: 2092-2099. 10.1007/s00330-010-1794-7.PubMedView ArticleGoogle Scholar
- Abbara S, Soni AV, Cury RC: Evaluation of cardiac function and valves by multidetector row computed tomography. Semin Roentgenol. 2008, 43: 145-153. 10.1053/j.ro.2008.01.009.PubMedView ArticleGoogle Scholar
- Park YJ, Kim YJ, Lee JW, Kim HY, Hong YJ, Lee HJ, Hur J, Nam JE, Choi BW: Automatic tube potential selection with tube current modulation (APSCM) in coronary CT angiography: comparison of image quality and radiation dose with conventional body mass index-based protocol. J Cardiovasc Comput Tomogr. 2012, 6: 184-190. 10.1016/j.jcct.2012.04.002.PubMedView ArticleGoogle Scholar
- Ghoshhajra BB, Engel LC, Karolyi M, Sidhu MS, Wai B, Barreto M, Shanmugam U, Hoffmann U, Brady TJ, Kalra M, Abbara S: Cardiac computed tomography angiography with automatic tube potential selection: effects on radiation dose and image quality. J Thorac Imaging. 2012, 28 (1): 40-48.View ArticleGoogle Scholar
- Jabbour A, Ismail TF, Moat N, Gulati A, Roussin I, Alpendurada F, Park B, Okoroafor F, Asgar A, Barker S: Multimodality imaging in transcatheter aortic valve implantation and post-procedural aortic regurgitation: comparison among cardiovascular magnetic resonance, cardiac computed tomography, and echocardiography. J Am Coll Cardiol. 2011, 58: 2165-2173. 10.1016/j.jacc.2011.09.010.PubMedView ArticleGoogle Scholar
- Techasith T, Ghoshhajra BB, Truong QA, Pale R, Nasir K, Bolen MA, Hoffmann U, Cury RC, Abbara S, Brady TJ, Blankstein R: The effect of heart rhythm on patient radiation dose with dual-source cardiac computed tomography. J Cardiovasc Comput Tomogr. 2011, 5: 255-263. 10.1016/j.jcct.2011.05.003.PubMedView ArticleGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.