- Research article
- Open Access
Effect of treatment with an overheated dry-saturated steam vapour disinfection system on multidrug and extensively drug-resistant nosocomial pathogens and comparison with sodium hypochlorite activity
BMC Research Notes volume 8, Article number: 551 (2015)
The development of portable steam generators has made disinfection of the environment more practical. This study assessed the “in vitro” ability of an overheated dry-saturated steam vapour system to kill multidrug and extensively-drug resistant nosocomial pathogens, defining the antimicrobial spectrum and the contact times compared with the activity of sodium hypochlorite.
The antibacterial efficacy of the overheated dry-saturated steam vapour system and of sodium hypochlorite against nosocomial pathogen isolates: extensively drug-resistant Acinetobacter baumannii, Pseudomonas aeruginosa, carbapenemase-producing Klebsiella pneumoniae, methicillin-resistant Staphylococcus aureus, high-level aminoglycoside-resistant Enterococcus faecalis, Candida parapsilosis and Aspergillus fumigatus were assessed using a surface time-kill test carried out on glass surfaces, with or without bovine serum albumin (BSA).
The bactericidal activity of the overheated dry-saturated steam vapour system was observed at 180 °C after 5 min contact with or without BSA, using an initial inoculum of 109 CFU/mL. To reduce C. parapsilosis and A. fumigatus counts (from 107 CFU/mL), a longer contact time was necessary (7 min). In vitro tests with sodium hypochlorite at 5 % in the absence of an organic substance also resulted in an overall reduction in bacterial counts (from 109 CFU/mL) after 5 min of treatment. For mycotic challenge (107 CFU/mL), a longer contact time was necessary (7 min). In the presence of an organic substance, after 5 min, the hypochlorite reduced the viable count from 109 to 105 CFU/mL for all bacterial strains except E. faecalis that showed a reduction of 2 log units (109 to 107 CFU/mL). For C. parapsilosis and A. fumigatus, a 2 log unit reduction was observed after 7 min.
Steam disinfection of environmental surfaces using a portable steam generator is a practical and effective method that is not affected by the presence of organic matter.
Recent studies have demonstrated that several major nosocomial pathogens are shed by patients and contaminate environmental surfaces at concentrations sufficient for transmission [1–3]. Such pathogens can survive for extended periods despite cleaning with chlorine-releasing disinfectants [4, 5] and can be transferred to the hands of healthcare workers . The spread of nosocomial pathogens has been linked to poor hand-hygiene practices. However, healthcare workers are more likely to contaminate their hands from touching the patient environment than from patient contact .
Mounting evidence demonstrates that outbreak strains of methicillin-resistant Staphylococcus aureus (MRSA), extensively drug-resistant (XDR) Acinetobacter baumannii, Pseudomonas aeruginosa and (extended spectrum beta-lactamase)-producing Enterobacteriaceae survived significantly longer on environmental surfaces than non-outbreak strains, indicating a possible fitness advantage [7–13].
In healthcare settings, surfaces are usually decontaminated using liquid chemical disinfectants, often chlorine derivatives . However, these products have some drawbacks, they are usually toxic to humans, they display chemical reactivity, and they require long periods of contact (up to 15 min) with surfaces to kill microorganisms . Moreover, some nosocomial pathogens are resistant to many disinfectants [16–18].
The development of portable steam generators has made the disinfection of environments more practical [19–21]. The aim of the present study was to assess “in vitro” the ability of the overheated dry-saturated steam vapour system to kill multidrug and XDR nosocomial pathogens on surfaces, and to define the antimicrobial spectrum and the contact times required by this system. Our results were then compared with those obtained using sodium hypochlorite, an agent commonly used in clinical sanitation procedures [14, 15].
The in vitro tests were carried out in a microbiology laboratory. Glass surfaces were chosen because they are flat, inert, easy to contaminate and highly resistant to chemical products and to heat.
The dimensions of the surfaces were 50 × 50 cm, with a thickness of 30 mm and a weight of 38 kg/m2.
Disinfection using the overheated dry-saturated steam vapour system
The steam generator device consists of a professional steam generator (Sani System Polti, Medical Division Polti s.p.a., Como, Italy) (Fig. 1). The dimensions of the unit were 47 × 45 × 90.5 cm, with a weight of 27.5 kg. The portable unit was outfitted with a hose connected to a steam dispenser. The overheated dry-saturated steam vapour was high temperature steam generated in the steel boiler that reached a pressure of 6 bar and was then further superheated in an expansion chamber to generate a dry saturated steam vapour at 180 °C. The unit was filled with tap water. Fifteen minutes before use, the unit was activated to reach the maximum operating boiler pressure (6 bar) in accordance with the manufacturer’s instructions (Sani System Polti). Protection equipment, such as heat-resistant gloves, safety aprons or glasses, is not required by workers when operating this system.
Disinfection with sodium hypochlorite
The commercial product Decs containing sodium hypochlorite at 2.8 % (2.7 % active chlorine) (Lombarda H s.r.l, Albairate, Milan, Italy) was used in this study. The stock solution of the product was diluted in sterilised water to a final concentration of 5 % to obtain approximately 1400 ppm active chlorine. This is the concentration of sodium hypochlorite solution usually used for disinfection in hospitals .
The bactericidal effects of the overheated dry-saturated steam vapour and of sodium hypochlorite were evaluated using seven environmental organisms, including Gram-positive and Gram-negative bacteria, yeast and fungi: XDR A. baumannii (strain 4500/2010), P. aeruginosa (strain 3637/2006), MRSA (strain 3582/2006), high-level aminoglycoside-resistant (HLAR) Enterococcus faecalis (strain 3084/2005), carbapenemase-producing Klebsiella pneumoniae (KPC; strain 4640/2012), Candida parapsilosis (strain 4093/2009) and Aspergillus fumigatus (strain 3430/2006). These environmental strains were isolated between January 2005 and December 2012 in the neonatal and the adult intensive care units of the University Hospital “Federico II” in Naples, Italy, during environmental microbiological investigations performed to identify sources and reservoirs of infection in the course of nosocomial outbreaks [10, 11].
Environmental isolates were identified by commercial systems (VITEK® 2 automatic system; bioMèrieux Marcy-L’Etoile, France and Becton–Dickinson Phoenix, Phoenix Technologies Ltd, San Jose, CA, USA). All isolates were stored at −80 °C in glycerol solution.
Susceptibility testing and screening
The strains used in our study were selected because of their antimicrobial resistance phenotypes.
Antimicrobial susceptibility patterns were analysed using an automated system (BD Phoenix) and by manual methods (i.e., Kirby-Bauer disk diffusion assay, Etest and microdilution tests) and the results were interpreted according to EUCAST . Antimicrobial susceptibility testing revealed a multidrug-resistant antibiotype for all of the isolated microorganisms. The antimicrobial susceptibility patterns of strains included in the study are listed in Tables 1 and 2.
In vitro time-kill tests
The bactericidal effects of the overheated dry-saturated steam vapour and the sodium hypochlorite on multidrug-resistant nosocomial pathogens were assessed by measuring viable cell counts using the quantitative time-kill test  as recommended by the European Committee for Standardization (CEN EN1276) with or without bovine serum albumin (BSA 0.3 g/100 mL) . The quantitative time-kill test was performed as previously described . In brief, a logarithmic-phase culture was adjusted to ca. 109 CFU/mL for bacteria and 107 CFU/mL for yeasts and fungi in Luria–Bertani broth. The test was carried out on glass surfaces initially contaminated with 100 μL of microbial suspension, prepared as above, with and without BSA. The surfaces were sanitised with the overheated dry-saturated steam vapour generated at 180 °C at different time points (1, 2, 3 min, up to 8 min). At each time point, the surfaces were rinsed with 1 mL of a sterile physiological solution, and 100 μL was removed from the rinsing solution and serially diluted (1:10, 1:100, 1:1000, 1:10,000) in phosphate-buffered saline (pH 7.4) and spread on Bacto D/E neutralizing agar (Becton–Dickinson). Plates were incubated at 37 °C for 24 h for bacterial strains and at 32 °C for 72 h for C. parapsilosis and A. fumigatus strains. Each experiment was performed in triplicate.
Time-kill tests following hypochlorite disinfection were carried out on glass surfaces with and without BSA. The glass surfaces were contaminated at eight different points with 100 μL of each microbial suspension. The surfaces were sanitised at each inoculum point with 900 µL of disinfectant solution at room temperature (22 °C) and then 100 µL of each suspension was removed from the surfaces at different time points and spread on Bacto D/E neutralizing agar after serial dilution, as described above. Plates were incubated under the same conditions as above. The antimicrobial action of chlorine was neutralised by sodium thiosulphate , which is contained within the D/E neutralizing agar at a concentration of 6.0 g/L. Each experiment was performed in triplicate.
Viable cell counts of the bacterial, yeast and fungal strains were evaluated on agar plates (Bacto D/E neutralizing agar) after incubation under the conditions described above. The results of the in vitro tests were interpreted in accordance with the CEN standards . According to these standards, the bactericidal activity of disinfectant was defined as a ≥4 log 10 CFU/mL decrease in the viable count compared with the initial inoculum.
Data were analysed using GraphPad Prism v.5.04 software (GraphPad Software, La Jolla, CA, USA). The significance of the reduction in viable counts was analysed using the Student’s t test and the reduction was considered significant when the P value was <0.05.
Results and discussion
Overheated dry-saturated steam vapour
The bactericidal activity of the dry saturated, steam vapour system was observed for Gram-negative bacteria at a temperature of 180 °C after 5 min of treatment without BSA. A significant reduction in P. aeruginosa (p = 0.0004, as determined by the Student’s t test compared with untreated samples), XDR A. baumannii (p < 0.0001) and KPC (p < 0.0001), from an initial concentration of 109 CFU/mL, was observed after 2 min of treatment with this system. The same results were obtained in the presence of BSA (Fig. 2). For MRSA and HLAR E. faecalis, bactericidal activity was observed after 5 min contact time with a significant reduction (p < 0.0001) from the initial concentration (109 CFU/mL) after 3 min of treatment, in the presence and absence of BSA (Fig. 2).
A longer contact time (7 min) was necessary to reduce the viable count of C. parapsilosis and A. fumigatus (to 107 CFU/mL) without or with BSA, although a significant reduction in the yeast (p < 0.0001) and fungi (p = 0.0004) counts from the initial concentration (107 CFU/mL) was observed after 3 min of treatment (Fig. 2).
The in vitro tests with sodium hypochlorite at 5 % in the absence of an organic substance (i.e. BSA) resulted in an overall reduction in the viable count within 5 min for Gram-negative and Gram-positive bacteria (from 109 CFU/mL) (Fig. 3). A longer contact time (7 min) was necessary under the same conditions without BSA to reduce the yeast and fungi count (from 107 CFU/mL) (Fig. 3).
Conversely, 5-min hypochlorite treatment in the presence of an organic substance reduced the initial viable count (109 CFU/mL) to 105 CFU/mL (4 log decrease) for all bacterial strains, except HLAR E. faecalis that showed a 2 log decrease (109 to 107 CFU/mL). The reduction in C. parapsilosis and A. fumigatus counts (107 to 105 CFU/mL) at 7 min was only 2 log units in the presence of BSA (Fig. 3).
The disinfection of all multidrug-resistant pathogens was rapid and complete. This is an important result because surface-related transmission of antibiotic-resistant microorganisms is a growing threat in healthcare settings and chemical disinfectants may have the potential to select cross-resistance to antibiotics [16, 27].
The in vitro tests using sodium hypochlorite at 5 % in the absence of an organic substance also resulted in an overall reduction in bacterial and mycotic concentrations. In the presence of an organic substance, the antimicrobial activity of sodium hypochlorite was reduced. The use of disinfectants requires an initial cleaning step to remove the organic matter that would otherwise “consume” the oxidizing disinfectant [14, 15]. Therefore, in the presence of an organic substance, a large quantity of oxidising disinfectant (containing chlorine and ozone) may be required, often at higher-than-standard concentrations. The disinfecting effect of chlorine is only visible when the “requirement” of organic substances has been met [14, 15]. In contrast, the steam vapour device depends on heat for efficacy and the presence of organic matter does not influence the effectiveness . Moist heat acts by denaturation and coagulation of protein, breakage of DNA strands and loss of functional integrity of cell membranes, resulting in cell death .
This system could help to reduce the risk of spreading nosocomial infections in healthcare facilities.
Our time-kill curve studies revealed the time-dependent effect of killing by steam vapour disinfection with an overall drop in microbial counts obtained at either 5 or 7 min depending on the microorganism. These findings are in accordance with previous studies performed using saturated steam showing that this method is effective in decontaminating surfaces contaminated with high concentrations of pathogenic microorganisms, killing 100 % of bacteria under all experimental conditions [19–21]. The antimicrobial activity of the steam disinfection system has great potential in the disinfection of contaminated hospital environments, because many of the most important nosocomial pathogens, such as P. aeruginosa and A. baumannii, have natural resistance to liquid chemical disinfectants and especially to quaternary ammonium compounds . The saturated steam vapour disinfection system has a broad range of activity [19–21] and this technology is also considered to have great potential for eliminating biofilms, aggregates of active cells embedded within a polymeric matrix and attached to a biotic or abiotic surface .
The steam vapour system has been proven to reduce or completely eliminate microbial contamination on hard surfaces at earlier and later time points, respectively. This portable device quickly reduced and then eliminated microbial loads in the presence or absence or an organic substance in contrast to sodium hypochlorite, whose disinfectant effect is only visible when the “requirement” of the organic substance has been met. The findings of this study suggest that the portable vapour disinfection system is a viable alternative to available chemical disinfectants, including chloride derivatives, for the disinfection of hospital environmental surfaces.
Weber DJ, Rutala WA, Miller MB, Huslage K, Sickbert-Bennett E. Role of Hospital surfaces in the transmission of emerging health care-associated pathogens: norovirus, Clostridium difficile and Acinetobacter species. Am J Infect Control. 2010;38(5 Suppl 1):S25–33.
Wagenvoort JH, De Brauwer EI, Penders RJ, Willems RJ, Top J, Bonten MJ. Environmental survival of vancomycin-resistant Enterococcus faecium. J Hosp Infect. 2011;77:282–3.
Coughenour C, Stevens V, Stetzenbach LD. An evaluation of methicillin-resistant Staphylococcus aureus survival on five environmental surfaces. Microb Drug Resist. 2011;17:457–61.
Kramer A, Shwebke I, Kampf G. How long do nosocomial pathogens persist on inanimate surfaces? A systematic review. BMC Infect Dis. 2006;6:130–8.
Webster C, Towner KJ, Humphreys H. Survival of Acinetobacter on three clinically related inanimate surfaces. Infect Control Hosp Epidemiol. 2000;2:246.
Stiefel U, Cadnum JL, Eckstein BC, Guerrero DM, Tima MA, Donskey CJ. Contamination of hands with methicillin-resistant Staphylococus aureus after contact with environmental surfaces and after contact with the skin of colonized patients. Infect Control Hosp Epidemiol. 2011;32:185–7.
Wagenvoort JH, Sluijsmans W, Penders RJ. Better environmental survival of outbreak vs sporadic MRSA isolates. J Hosp Infect. 2000;45:231–4.
Magiorakos AP, Srinivasan A, Carey RB, Carmeli Y, Falagas ME, Giske CG, Harbarth S, Hindler JF, Kahlmeter G, Olsson-Liljequist B, Paterson DL, Rice LB, Stelling J, Struelens MJ, Vatopoulos A, Weber JT, Monnet DL. Multidrug-resistant, extensively drug-resistant and pandrug-resistant bacteria: an international expert proposal for interim standard definitions for acquired resistance. Clin Microbiol Infect. 2012;13:268–81.
Gad GF, El-Domany RA, Zaki S, Ashour HM. Characterization of Pseudomonas aeruginosa isolated from clinical and environmental samples in Minia, Egypt: prevalence, antibiogram and resistance mechanisms. J Antimicrob Chemother. 2007;60:1010–7.
Zarrilli R, Di Popolo A, Bagattini M, Giannouli M, Martino D, Barchitta M, Quattrocchi A, Iula VD, de Luca C, Scarcella A, Triassi M, Agodi A. Clonal spread and patients’ risk factors for extensively drug-resistant Acinetobacter baumannii acquisition in a neonatal intensive care unit in Italy. J Hosp Inf. 2012;82:260–5.
Crivaro V, Di Popolo A, Caprio A, Lambiase A, Di Resta M, Borriello T, Scarcella A, Triassi M, Zarrilli R. Pseudomonas aeruginosa in a neonatal intensive care unit: molecular epidemiology and infection control measures. BMC Inf Dis. 2009;9:70.
Bagattini M, Crivaro V, Di Popolo A, Gentile F, Scarcella A, Triassi M, Villari P, Zarrilli R. Molecular epidemiology of extended-spectrum β-lactamase-producing Klebsiella pneumoniae in a neonatal intensive care unit. J Antimicrob Chemother. 2006;57:979–82.
Giannouli M, Antunes LCS, Marchetti V, Triassi M, Visca P, Zarrilli R. Virulence-related traits of epidemic Acinetobacter baumannii strains belonging to the international clonal lineages I-III and to the emerging genotypes ST25 and ST78. BMC Infect Dis. 2013;13:282. doi:10.1186/1471-2334-13-282.
Rutala WA, Weber DJ. Uses of inorganic hypoclorite (bleach) in health-care facilities. Clin Microb Rev. 1997;10:597–610.
Fukuzaki S. Mechanisms of actions of sodium hypochlorite in cleaning and disinfection processes. Biocontrol Sci. 2006;11:147–57.
McDonnell G, Denver Russell A. Antiseptics and disinfectants: activity, action and resistance. Clin Microbiol Rev. 1999;1:147–79.
Levy SB. Antibiotic and antiseptic resistance: impact on public health. Pediatr Infect Dis J. 2000;19(Suppl 10):120–2.
Sheldon AT Jr. Antiseptic “Resistance”: real or perceived threat? Clin Infect Dis. 2005;40:1650–6.
De Lorenzi S, Salvatorelli G, Finzi GF, Cugini P. Use of a steam generator for disinfection of hospital ward room surfaces. Br Microbiol Res J. 2012;2:228–32.
Storni E, Donati ME, Naldi S, Pederzoli A, Maso A. In vitro evaluation of bactericidal activity of an over-heated saturated dry atomized steam electro-medical device on inert contaminated surfaces. Ann Ig. 2011;23:347–54.
Tanner BD. Reduction in infection risk through treatment of microbially contaminated surfaces with a novel, portable, saturated steam vapor disinfection system. Am J Infect Control. 2009;37:20–7.
Rutala WA, Weber DJ. Healthcare infection control practices advisory committee (HICPAC). Guideline for Disinfection and Sterilization in Healthcare Facilities, Atlanta, GA: CDC. 2008. http://cdc.gov/ncidod/dhqp/pdf/guidelines/Disinfection_Nov_2008.pdf. Accessed 22 Nov 2013.
European Committee on Antimicrobial Susceptibility Testing. Breakpoint tables for interpretation of MICs and Zone diameters Version 3.1 (bacteria) and 6.1 (fungi). 2013. http://www.eucast.org/clinical_breakpoints Accessed 22 Nov 2013.
Giannouli M, Di Popolo A, Durante-Mangoni E, Bernardo M, Cuccurullo S, Amato G, Tripodi M-F, Triassi M, Utili R, Zarrilli R. Molecular epidemiology and mechanisms of rifampicin resistance in Acinetobacter baumannii isolates from Italy. Int J Antimicrob Agents. 2012;39:58–63.
European Committee for Standardization. Chemical disinfectants and antiseptics. Quantitative suspension test for the evaluation of bactericidal activity of chemical disinfectants and antiseptics used in food, industrial, domestic, and industrial areas. Test method and requirements (phase 2, step 1). European standard EN 1276. Brussels 1997.
Bansal N, Sinha I, Virdi JS. Virulence plasmid (pYV)-associated susceptibility of Yersinia enterocolitica to chlorine and heavy metals. J Appl Microbiol. 2000;89:663–7.
Chuanchuen R, Beinlich K, Hoang TT, Becher A, Karkhoff-Schweizer RR, Schweizer HP. Cross-resistance between triclosan and antibiotics in Pseudomonas aeruginosa is mediated by multidrug efflux pumps: exposure of a susceptible mutant strain to triclosan selects nfxB mutants overexpressing MexCD-OprJ. Antimicrob Agents Chemother. 2001;45:428–32.
Williams GJ, Denyer SP, Hosein IK, Hill DW, Maillard JY. Limitations of the efficacy of surface disinfection in the healthcare setting. Infect Control Hosp Epidemiol. 2009;30:570–3.
Hussain M. Physical control of microorganisms: boiling water. 2009. http://microbiologyon-line.blogspot.com/2009/08/pasteurization.html. Accessed 03 Mar 2015.
Adair FW, Geftic SC, Gelzer J. Resistance of P. aeruginosa to quaternary ammonium compounds. I Growth in belzalkonium chloride solution. Appl Microbiol. 1969;18:299–302.
Song L, Jianfeng W, Xi C. Biofilms on environmental surfaces: evaluation of the disinfection efficacy of a novel steam vapor system. Am J of Infect Control. 2012;40:926–30.
MB conceived and designed the study. RBuo, DM, RBel and NG performed the assays. MG and AN analysed and interpreted the results of time-kill tests. RZ and MT supervised the study. MB and RZ wrote the paper. All authors read and approved the final manuscript.
The authors would like to thank Professor V. Iaccarino from the Department of Diagnostic Imaging and Radiotherapy, Section of Vascular and Interventional Radiology AOU Federico II Napoli, for his enthusiastic support. The authors received no financial support to carry out this study.
The authors declare that they have no competing interests.
About this article
Cite this article
Bagattini, M., Buonocore, R., Giannouli, M. et al. Effect of treatment with an overheated dry-saturated steam vapour disinfection system on multidrug and extensively drug-resistant nosocomial pathogens and comparison with sodium hypochlorite activity. BMC Res Notes 8, 551 (2015). https://doi.org/10.1186/s13104-015-1534-9
- Steam vapour disinfection system
- Chlorine or hypochlorite disinfection
- Nosocomial pathogens
- Multidrug and extensively drug-resistant bacteria