- Technical Note
- Open Access
Performance of Mycobacterium Growth Indicator Tube BACTEC 960 with Lowenstein–Jensen method for diagnosis of Mycobacterium tuberculosis at Ethiopian National Tuberculosis Reference Laboratory, Addis Ababa, Ethiopia
BMC Research Notes volume 10, Article number: 181 (2017)
Bacteriological confirmed active case detection remains the corner stone for diagnosing tuberculosis. Non-radiometric liquid culture system Mycobacterium Growth Indicator Tube with automated interface had been recommended by expert groups in addition to conventional solid culture media such as Lowenstein–Jensen. However in high burden resource limited countries advanced non-radiometric based tuberculosis diagnostic methods such as MGIT 960 is limited. Therefore we have evaluated the performance of MGIT 960 system compared to LJ for recovery of Mycobacterium complex (MTBC) from clinical specimens.
A cross sectional study was conducted from a total of 908 samples between January 1st, 2013 to December 31st, 2014. Clinical specimens were processed following standard procedures and the final suspension was inoculated to MGIT tubes and LJ slant. Identification and confirmation of MTBC was done by ZN staining and SD Bioline test. Data was analyzed by SPSS version 20. The sensitivity, specificity, recovery rate and the average turnaround time to recover the organism was computed.
From a total of 908 clinical specimens processed using both LJ and BACTEC MGIT liquid culture methods the recovery rate for LJ and MGIT, for smear positive samples was 66.7% (74/111) and 87.4% (97/ 111) respectively while for smear negative samples was 13.4% (108/797) and 17.4% (139/797) for LJ and MGIT methods respectively. The overall recovery rate for MGIT is significantly higher than LJ methods [26% (236/908; vs. 20%, 182/908, P = 0.002)]. The average turnaround time for smear positive samples was 16 and 31 days for MGIT and LJ respectively. Turnaround time for smear negative samples was 20 and 36 days for MGIT and LJ respectively. The overall agreement between MGIT and LJ was fairly good with Kappa value of 0.59 (P < 0.001). In the present study the contamination rate for MGIT is higher than the LJ methods, 15 and 9.3% respectively.
The BACTEC MGIT liquid culture system has better MTBC recovery rate with shorter turnaround time for both smear positive and negative clinical specimens compared to Conventional LJ method. However, efforts should be made in order to reduce the high contamination rate in BACTEC MGIT system and to lesser extent to LJ methods.
Symptomatic active form of TB is the most crucial factor for transmission of infection. Even though competent immune system will limit the multiplication, some bacilli may remain dormant latently. Relatively, small proportion of exposed people develops TB disease at any course of their life; the probability of developing TB is much higher among people infected with HIV . Burden of tuberculosis in developing countries remain high and affecting the most productive young ages as a result of an enormous economic lose in the populations of most low- income countries of Asia and Africa which were evident from many reports .
Bacteriological confirmed active case detection remains the corner stone for diagnosis of Tuberculosis, However, among the most utilized diagnostics methods; TB smear microscopy is the most popular among all available methods in developing countries. In addition, most commonly used diagnostic technique such as LJ based myco-bacteriological culture and recently introduced TB molecular diagnostic technique with efficient turnaround time and specificity is also used for the diagnosis of active forms of TB in clinical settings .
The most popular LJ based myco-bacteriological culture TB diagnostic methods, Lowenstein–Jensen (LJ), is time consuming, taking up to 6–8 weeks to recover Mycobacteria or discriminate negative samples [4, 5].
Non-radiometric liquid culture methods called Mycobacterium Growth Indicator Tube (MGIT; Becton Dickinson) system have been introduced for better recovery of Mycobacterium . This MGIT 960 system is a fully automated, non-radiometric, and used to incubate and monitor 960 samples at a time with automated result-reporting system . The methods designed to shorten turnaround time with good recovery rate than conventional solid TB culture system . The MGIT 960 system has been in place of the national tuberculosis laboratories of Ethiopia that is sued along with the conventional LJ methods. However, there overall performance of MGIT is not known for large samples. Hence, we aimed to evaluate the performance of BACTEC MGIT 960 system in comparison with Lowenstein–Jensen culture methods in Ethiopian context hoping that the finding could supplement TB diagnostic methods in high TB burden resource poor countries.
Methods and materials
A cross sectional study was conducted from total 908 TB suspected patients who were referred for sputum culture testing at Ethiopian Public Health Institute (EPHI), National TB Reference laboratory from January 1st, 2013–December 31st, 2014.
Specimens were processed by the sodium hydroxide and N-acetyl-l-cysteine (NaOH/NALC) method and concentrated by centrifugation (3000 RPM for 15 min). The supernatant was discarded, and the sediment was re suspended with sterile phosphate buffer to a final volume of 2 ml. This suspension was then used for making smears for acid-fast staining and for inoculation in to BACTEC MGIT 960 culture tube (0.5 ml) and LJ slant (2–3 drops). In most cases, the specimens were processed on the day of collection. Those that could not be processed on the day of collection (i.e., received during the weekend) were stored at 2–8 °C for 2 days. Samples collected and received more than 3 days were rejected based on the laboratory rejection criteria and documented on sample rejection log book. Smears were examined for acid-fast bacilli (AFB). All AFB smears were stained by the Ziehl–Neelsen staining method and examined with a bright filed microscope following the standard guidelines of our laboratory.
Culture of Mycobacteria
BACTEC MGIT 960 system
The BACTEC MGIT 960 culture tube contains 7 ml Middle brook 7H10 broth base, in which an enrichment supplement containing oleic acid, albumin, dextrose, and catalase is added. In addition, an antibiotic mixture of polymyxin B, amphotericin B, nalidixic acid, trimethoprim, and azlocillin is added to the culture tube to inhibit growth of other microbes.
The culture tube contains a fluorescent sensor that detects the concentration of oxygen in the culture medium. The level of fluorescence corresponds to the amount of oxygen consumed by the organisms in the inoculated specimens. This, in turn, is proportional to the number of bacteria present. When a certain level of fluorescence is reached, the instrument indicated that the tube is positive. After inoculation of each tube with 0.5 ml of the processed specimen, the tubes were incubated at 37 °C in the BACTEC MGIT 960 instrument and monitored automatically every 60 min for increased fluorescence. The culture tubes were maintained until it became positive or for 42 days of maximum for negative samples. Samples found to be positive were removed from the instrument then sub cultured on Brain Heart Infusion agar plate, incubated for 48 h at 37 °C to asses possible contaminants. All the positive tubes were further confirmed by ZN staining methods and further confirmed by MPT64 protein Specific detection immune chromatographic test (SD Bioline Kit, Standard Diagnostics, Inc., Korea). The kit can discriminate MTBC and non Mycobacterium tuberculosis complex.
All specimens which were processed were also inoculated onto conventional solid media. LJ slant. The LJ slant medium was considered positive upon appearance of colonies on the surface, and the time to detection was based on the earliest date of detection of colonies on the solid media, ultimately confirmed by positive AFB smears and MPT64 protein specific detection immuno chromatographic test (SD Bioline Kit, Standard Diagnostics, Inc., Korea). Solid media were incubated at 37 °C for 8 weeks, and were inspected twice weekly or until Mycobacterium colonies were seen.
A total of 908 patients samples were included for this evaluation purpose. In this study, 423 (46.6%) were new patients refereed to our laboratory and 485 (53.4%) patients were follow up patients that are refereed for Rifampicin and Isoniazid Resistance testing (Table 1).
The overall recovery rate of MTBC by BACTTEC MGIT 960 method was 14.4% (131/908) and 11.5% (105/908) for new TB patients and for MDR-TB follow up patients respectively. On the other hand, the conventional LJ methods recovered relatively lower no. MTBC cases, 11.3% (103/908) and 8.7% (79/908) respectively for new and MDR TB follow up patients.
Overall, 26% of MTBC was detected by BACTEC MGIT methods while LJ methods detected 20% (182/908) of MTBC cases. A total of 534 (58.8%) specimens were negative by either LJ or MGIT methods and 138 (15%) specimens were contaminated (Table 2). Relatively good agreement has been observed between BACTEC MGIT 960 and LJ methods with Kappa value of 0.59.
In this study, 139 (17.4%) and 108 (13.6%) of smear negative cases were found to be positive for MTBC using BACTEC MGIT 960 and LJ culture methods respectively. The recovery rate of BACTEC MGIT was found significantly higher than LJ method (P < 0.007). The overall agreement for MTBC detection between BACTEC MGIT 960 method and AFB smears method were 87.4%. Whereas the agreement between LJ culture methods and AFB smears were low, only 66.7% (Table 3).
BACTEC MGIT 960 system has higher contamination rate than LJ methods, 15% and 9.2 respectively. The contamination rate of MGIT 960 system and LJ method for new patients were 6.7 and 4% respectively (Table 4).
The average time to detect (TTD) significant growth of Mycobcateria from smear-positive specimens using BACTEC MGIT 960 was 16 days and 20 days for smear-negative patients. Whereas TTD using LJ method was 31 days for smear positive cases and 36 days for smear negative specimens. There is significant difference between culture methods and TTD, MGIT picked MTBC cases faster than LJ methods both for smear positive and smear negative cases (P < 0.001).
Non radiometric Mycobacterium Growth Indicator Tube Liquid culture is one of TB culture method which is endorsed and recommended methods by World Health Organization. The method is found to be rapid and has 10% improved sensitivity for recovery of MTBC over the conventional egg based LJ method. However, its utilization is limited to only in few high burden resources limited countries including Ethiopia. Hence in order to enhance the applicability of this method under resource limited settings, the present study was evaluated and significant difference has been observed between the BACTEC MGIT 960 and LJ methods interims of recovery rate and time of detection.
Improved recovery rate of Mycobacterium species were observed compared to the conventional LJ method with significant shorter turnaround time (TAT) for both smear positive and Negative sputum specimens (87.4 vs. 66.7% for MGIT and LJ methods respectively). This finding is comparable with similar study from Nigeria for smear positive specimen using BACTEC MGIT 960 system with recovery rate of 87% but improved recovery rate were reported using LJ method with the rate of 78% than our finding . Comparable findings were reported in USA with the rate of 82% for BACTEC MGIT and 76% with LJ method . South African study reported almost similar finding with recovery rate of 83% using BACTEC MGIT for smear positive specimens .
Higher recovery rates were reported in Taiwan from smear positive specimen with 100% using BACTEC MGIT 960 system and 92.9% with LJ method . Similarly, higher recovery rates were reported from Germany, Turkey, Yugoslavia, Hungary and Spain from smear positive specimen with rates of 94.7% and 94.7, 88 and 83%, 93.2 and 67.3%, 96.4 and 81.8%, 95.5 and 79.5% for BACTEC MGIT and LJ methods respectively [14,15,16,17]. Unlike the present study, a much higher recovery rate was reported from Iraq using similar platform with the rate of 100% for MGIT 960 and 72.6% for LJ method . A study from Portland, Oregon also reported a recovery rate of 100 and 90.3% for BACTEC MGIT 960 AND LJ method respectively . Another studies from Pakistan, also reported higher recovery rate of 97.6 and 83.7% for BACTEC MGIT and LJ respectively . On the other hand, lower recovery rate was reported from USA and Hungary with the rate of 85.5 and 59.7%, 81 and 64.3%, for BACTEC MGIT and LJ methods respectively [8, 21]. The most likely difference between our findings with the aforementioned report could be difference in sample size, burden of tuberculosis and the sample size.
In this study the recovery rate for smear negative samples using BACTEC MGIT method were 17. 4% (139/797) compared 13.6% (108/797) for LJ method. This observation is in consistent with the findings of previous study in Nigeria that reported a recovery rate of 17 and 11% for MGIT and LJ method respectively .
Our finding is lower than report from smear negative Iraqi patients, with recovery rate of 24.2 and 14.7% for BACTEC MGIT and LJ method respectively . Australian workers reported a recovery rate of 21.8 and 19.4%  for MGIT and LJ method respectively.
Relatively lower recovery rate from smear negative patients was reported from USA and Hungary with the rate of 14.3 and 10%, 16.6 and 6.6% for BACTEC MGIT and LJ respectively [8, 21]. The recovery rate difference from smear positive and negative specimen might be associated with number of viable AFB inoculated into MGIT tube, the type of species of Mycobacteria, such as M. tuberculosis and Mycobacterium bovis that grow more slowly than NTM, such as Mycobacterium avium complex. It also depends on certain types of specimen quality. Another important factor that can lead to variable growth rate was the treatment status of patients. Particularly Specimens from chronically treated patients with drug resistant TB take a longer time to grow. This is true in our case, were more than 50% of the specimens were processed for MDR TB follow up patients.
Irrespective of the pre analytical variable mentioned so far, procedure and methodology used, potentially affects growth rate. Especially high pH or very low pH may cause injury or death to Mycobacteria during processing of the specimen. As a result it takes longer TAT for revival and growth of viable Mycobacteria. Some researcher reported that as many as 60–70% of the Mycobacteria are killed during sputum processing and centrifugation [18, 28, 32].
Another key feature of a culture method is contamination rate, in our study we found high contamination rate compared to pre-established standard of 5–8% for liquid culture and 3–5% for solid LJ culture methods , according to our finding, both BACTEC MGIT 960 and LJ methods had high contamination rates (15 and 9.3% respectively) . Similar findings has been reported from U.S.A, Taiwan, Spain and Turkey under similar evaluation conditions with contamination rates of 17, 15.1, 13.6, 12, 9% for the BACTEC MGIT method respectively [6, 22,23,24,25]. Unlike our study findings acceptable contamination rate were reported from Nigeria, Iraq and Germany with the rate of 7, 4.8 and 8.1% respectively for BACTEC MGIT method [11, 19, 26]. The discrepancy could be explained due to problems in sample collection, transportation, patient instructions, rejection criteria of the laboratory and difference in sample storage conditions .
Average time to detect minimum positive growth unit (70 Colony Forming Units of M. tuberculosis) using BACTEC MGIT 960 from smear-positive specimens were 16 days compared to 31 days for manual LJ methods. Comparable TTD for smear positive specimen was reported from Yugoslavia and India with average TTD time for BACTEC MGIT and LJ method of 13.7 days and LJ 22.1 days; 13.1 and 23.9 days respectively [13, 16]. Shorter recovery days were reported by Luqman Satti et al. from Pakistan with TTD of 11.2 days for BACTEC MGIT system . Similarly, shorter TTD was also reported from Taiwan and Turkey with average TTD days of 9.1 and 17.6 for BACTEC MGIT and 8.3 and 20.1 days for LJ methods [6, 25].
In our study a mean TTD, 20 days was recorded using BACTEC MGIT and 36 days using LJ method for smear negative patients Comparable findings were also reported by Piersimoni et al. in Malaysian and Italian researchers with TTD for MGIT/ LJ of 20/33.1, 13.2/35.3 and 13.3/25.6 days respectively [27, 29, 30]. The variation between studies could be due to the nature of patient categories and other factors such as low bacillary load from treatment groups .
In this study 13 (1.4%) and 33 (3.6%) specimens were smear-positive but culture-negative based on BACTEC MGIT and LJ methods respectively. Higher rate of 3.2 and 5.4% were reported for BACTEC MGIT and LJ method respectively ; some study also reported 6% each for BACTEC MGIT and LJ method . Very high rates were reported for smear positive cases with the rate of 6.5% by BACTEC MGIT and 9.2% by LJ methods . This discrepancy might be associated with factor such as harsh method used to decontaminate the specimen or prolonged exposure of decontaminant, NaOH during processing . It could be also associated with media quality and composition . Additional factors like good laboratory practice to follow and inspect the quality control methods for every batch of culture media and specimen may also contribute for differences .
We have observed that the overall performance of BACTEC MGIT liquid culture system is found to be better than the conventional LJ methods for rapid recovery of M. tuberculosis complex with shorter turnaround time for both smear positive and negative clinical specimens. However in efforts should be made in order to minimize the contamination rate of both BACTEC MGIT and LJ methods. Contentious improvement plan is required for provision of quality TB laboratory services which are crucial for control and eradication of tuberculosis at large.
drug sensitivity testing
Ethiopian Public Health Institute
Mycobacterium Growth Indicator Tube
Mycobacterium tuberculosis complex
National Tuberculosis Reference Laboratory
revolution per minutes
time to detection
Ndungu P, Revathi G, Kariuki S, Nganga Z. Risk factors in the transmission of tuberculosis in Nairobi: a descriptive epidemiological study. Sci Res Adv Microbiol. 2013;3:160–5.
Global tuberculosis control report. Geneva: WHO; 2013.
Cheng VC, Yew WW, Yuen KY. Molecular diagnostics in tuberculosis. Eur J Clin Microbiol Infect Dis. 2005;24:711–20.
Negi SS, Khan SF, Gupta S, Pasha ST, Khare S, Lal S. Comparison of the conventional diagnostic modalities, bactec culture and polymerase chain reaction test for diagnosis of tuberculosis. Indian J Med Microbiol. 2005;23:29–33.
Kent PT, Kubika GP. Public health mycobacteriology: a guide for a level III laboratory. Centers for Disease Control, Atlanta, Ga; 1985.
Lee J, Sou J, Lin CB, Wang JD, Lin TY, Tsai YC, et al. Comparative evaluation of the BACTEC MGIT 960 system with solid medium for identification of Mycobacterium. Int J Tuberc Lung Dis. 2003;7(6):569–74.
Abdel-Aziz N, Ahmed H, Morsy MM, Sabet E. A comparative evaluation of the BACTEC MGIT 960 (Mycobacterium Growth Indicator Tube) system with LJ solid medium for diagnosis of pulmonary tuberculosis. Egypt J Med Microbiol. 2009;18(2):91–6.
Lu D, Heeren B, Dunne W. Comparison of the automated Mycobacterium Growth Indicator Tube system (BACTEC 960/MGIT) with Lowenstein–Jensen medium for recovery of Mycobacterium from clinical specimens. J Am Soc Clin Pathol. 2002;118:542–5.
Lawson L, Emenyonu N, Abdurrahman ST, Lawson JO, Uzoewulu GN, Sogaolu MO, et al. Comparison of Mycobacterium tuberculosis drug susceptibility using solid and liquid culture in Nigeria. BMC Res Notes. 2013;6(215):1–5.
Badak FZ, Kiska DL, Setterquist S, Hartley C, O’connell MA, Hopfer RL, et al. Comparison of Mycobacteria Growth Indicator Tube with BACTEC for detection and recovery of Mycobacteria from clinical specimens. J Clin Microbiol. 1996;34(9):2236–9.
Brittle W, Marais BJ, Hesseling AC, Schaaf HS, Kidd M, Wasserman E, et al. Improvement in Mycobacterial yield and reduced time to detection in pediatric samples by use of a nutrient broth growth supplement. J Clin Microbiol. 2009;47(5):1287–9.
Satti L, Ikram A, Abbasi S, Butt T, Malik N, Mirza IA. Evaluation of BACTEC MGIT 960 system for recovery of Mycobacterium tuberculosis complex in Pakistan. Malays J Microbiol. 2010;6(2):203–8.
Sewell SL, Rashad AL, Rourke WJ, Sharon JR, Poor SL, Mccarthy JA, et al. Comparison of the Septi-Chek AFB and BACTEC systems and conventional culture for recovery of Mycobacteria. J Clin Microbiol. 1993;31(10):2689–91.
Leitritz L, Schubert SR, Cherl BB, Masch A, Heesemann JR, Roggenkamp A, et al. Evaluation of BACTEC MGIT 960 and BACTEC 460 TB systems for recovery of Mycobacteria from clinical specimens of a university hospital with low incidence of tuberculosis. J Clin Microbiol. 2001;39(10):3764–7.
Kunduracioglu A, Karasu I, Bicmen C, Ozsoz A, Erbaycu AE. Comparison of performances of MTD Gene-Probe® Test, BACTEC 960™ system and Lowenstein–Jensen culture methods in the diagnosis of smear-negative tuberculosis cases. Mikrobiyol Bul. 2013;47(3):417–31.
Mirovic V, Lepsanovic Z. Evaluation of the MB/BacT system for recovery of Mycobacteria from clinical specimens in comparison to Lowenstein–Jensen medium. Clin Microbiol Infect. 2002;8:709–14.
Somosko KA, Kodmon C, Lantos A, Bartfai Z, Tamasi L, Fuzy J, et al. Comparison of recoveries of Mycobacterium tuberculosis using the automated BACTEC MGIT 960 system, the BACTEC 460 TB system, and Lowenstein–Jensen medium. J Clin Microbiol. 2000;38(3):2395–7.
Kent PT, Kubica GP. The sputum digestion process in mycobacteriology: centrifugation efficacy and digestant toxicity. Annual Meeting of American Society for Microbiology, St. Louis, MO; 1984.
AL-Mazini Mo A, Bukeet T, Abdul Kareem A. Comparison of recoveries of Mycobacterium tuberculosis using the automated BACTEC MGIT 960 system and Lowenstein–Jensen medium. AL Qadisiya J Vet Med Sci. 2010;9(1):2395.
Anargyros P, Astill DS, Lim IS. Comparison of improved BACTEC and Lowenstein–Jensen media for culture of Mycobacterium from clinical specimens. J Clin Microbiol. 1990;28(6):1288–91.
Somoskovi Á, Magyar P. Comparison of the Mycobacterium Growth Indicator Tube with MB Redox, Lowenstein–Jensen, and Middle brook 7H11 media for recovery of Mycobacterium in clinical specimens. J Clin Microbiol. 1999;37(5):1366–9.
Phyffer GE, Welscher HM, Kissling P, Cieslak C, Casal ML, Gutierrez J, et al. Comparison of Mycobacteria Growth Indicator Tube (MGIT) with radiometric and solid culture for recovery of acid-fast bacilli. J Clin Microbiol. 1997;35(2):364–8.
Williams-Bouyer N, Yorke R, Lee HI, et al. Comparison of the BACTEC MGIT 960 and ESP culture system II for growth and detection of Mycobacteria. J Clin Microbiol. 2000;38:4167–70.
Otu J, Antonio M, Cheung YB, Donkor S, De Jong BC, Corrah T, et al. Comparative evaluation of BACTEC MGIT 960 with BACTEC 9000 MB and LJ for isolation of Mycobacterium in The Gambia. J Infect Dev Ctries. 2008;2(3):200–5.
Feyzioglu B, Dogan M, Sanli OO, Ozdemir M, Baykan M. Comparison of the performance of TK system with LJ and MGIT methods in the diagnosis of tuberculosis. Int J Clin Exp Med. 2014;7(4):1084–8.
Hanna BA, Ebrahimzadeh A, Elliott B, Morgan MA, Novak SM, Rusch-gerdes S, et al. Multicenter evaluation of the BACTEC MGIT 960 system for recovery of Mycobacteria. J Clin Microbiol. 1999;37(3):748–52.
Piersimoni C, Scarparo C, Callegaro A, Passerine C, Nista D, Bornigia S, et al. Comparison of MB/BacT ALERT 3D system with radiometric BACTEC system and Lowenstein–Jensen medium for recovery and identification of Mycobacteria from clinical specimens. J Clin Microbiol. 2001;39(2):651–7.
Siddiqi SH, Rüsch-Gerdes S. MGIT procedure manual. Foundation for innovative new diagnostics. Geneva, Switzerland; 2006.
Tortoli E, Cichero P, Piersimoni C, Simonetti MT, Gesu G, Nista D, et al. Use of BACTEC MGIT 960 for recovery of Mycobacteria from clinical specimens: multicenter study. J Clin Microbiol. 1999;37(11):3578–82.
Mohd F, Kee Peng NG, Yun Fong N. The manual MGIT system for the detection of M. tuberculosis in respiratory specimens: an experience in the University Malaya Medical Centre. Malays J Pathol. 2009;31(2):93–7.
Chien HP, Yu MC, Wu MH, Lin TP, Luh KT, et al. Comparison of the BACTEC MGIT 960 with Lowenstein–Jensen medium for recovery of Mycobacterium from clinical specimens. Int J Tuberc Lung Dis. 2000;4(9):866–70.
Isenberg HD. Clinical microbiology procedure handbook. Washington, DC: American Society for Microbiology; 2004.
GD, AK, ZY, MG, MT involved in all designing the manuscript, writing and analysis, KD, GD, KG, AM, ZD, JD, MH contributed in methodological advice, editing and reviewing and, SM, MH, ZD, AM contributed in technical advice and analysis of the data. All authors read the manuscript before submission. All authors read and approved the final manuscript.
We acknowledged Ethiopian Public Health Institute and Department of Medical Laboratory Sciences, CHS, Addis Ababa University. Moreover, we are very grateful for patients who were participating in this study.
The authors declare that they have no competing interests.
Availability of data and materials
The data related with this manuscript will only can be found with communication with the authors with the following link email@example.com.
Consent for publications
We declare that, our manuscript doesn’t contain any image or individuals data.
Ethics approval and consent to participate
This research was conducted after obtaining ethical clearance from Addis Ababa University, College of Health Sciences, Department of Medical Laboratory Sciences and Ethiopian Public Health institute research and Ethics review board. Consent was also obtained from each study participant. Moreover, results were communicated to the requesting institution. All patients’ data were kept confidential.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Diriba, G., Kebede, A., Yaregal, Z. et al. Performance of Mycobacterium Growth Indicator Tube BACTEC 960 with Lowenstein–Jensen method for diagnosis of Mycobacterium tuberculosis at Ethiopian National Tuberculosis Reference Laboratory, Addis Ababa, Ethiopia. BMC Res Notes 10, 181 (2017). https://doi.org/10.1186/s13104-017-2497-9
- Mycobacteria Growth Indicator Tube