- Research note
- Open Access
Spectrophotometry assays to determine G6PD activity from Trinity Biotech and Pointe Scientific G6PD show good correlation
BMC Research Notes volume 11, Article number: 855 (2018)
Spectrophotometry kits from Pointe Scientific (PS; USA) were compared to kits from Trinity Biotech (Trinity; Ireland) in 50 venous blood samples from purposively selected individuals in Bangladesh. Repeatability and inter-assay variability were assessed by Students t-test, Bland-Altman plot and Pearson correlation coefficient (r). The median glucose-6-phosphate dehydrogenase (G6PD) activity of all G6PD normal participants was calculated per assay and defined as 100% activity. Performance was calculated considering 30% and 70% cut off activities and Trinity as reference.
The intra-assay correlation of Trinity (r = 0.9841, p < 0.001) and PS (r = 0.9833, p < 0.001) did not differ significantly (p = 0.904). Both assays were closely correlated (r = 0.9799, p < 0.001), with a mean difference of 0.1 U/gHb (95% limit of agreement: − 1.32 to 1.57). At 30% cut off PS had a sensitivity of 100% (95% confidence interval (95 CI) 59.0–100.0) and specificity of 100% (95% CI 91.8 to 100.0), at 70% cut-off of 100% (95% CI 79.4–100.0) and 97.1% (95% CI 84.7–99.9) respectively. The G6PD assay from PS is a reliable alternative to the assay from Trinity.
Glucose-6-phosphate dehydrogenase (G6PD) is the rate limiting step of the pentose-phosphate-pathway (PPP), which reduces nicotinamide adenine dinucleotide phosphate (NADP+) to nicotinamide adenine dinucleotide phosphate hydrogen (NADPH) . G6PD deficiency is the most common enzymopathy worldwide, affecting approximately 400 million people globally . G6PD deficiency does not affect life expectancy in most cases it is however a known risk factor for hyperbilirubinemia and kernicterus  and can result in hemolysis induced by specific compounds such as fava beans, sulphonamides, quinolones, dapsone and the 8-aminoquinoline class of antimalarials .
The reference method for the quantification of G6PD enzyme activity is spectrophotometry [4,5,6], a method based on the colorimetric detection of NADPH. In brief a defined amount of hemolysate is added to a solution containing glucose-6-phosphate and NADP, which results in the formation of NADPH facilitated through G6PD. Depending on the activity of the G6PD, NADPH production varies and can be measured at a wavelength of 340 nm over a predefined time interval at standardized temperatures . There are a number of spectrophotometry kits from different providers available, one of the most common is that manufactured by Trinity Biotech (Trinity; Ireland). In 2017 the company suspended their production due to difficulties in maintaining quality standards for their reagents. Pointe Scientific (PS; USA) also produce reagents for G6PD spectrophotometry that are similar to Trinity in their operational characteristics, however are designed for a higher optimal measuring temperature. The aim of this study was to compare the intra and inter assay variability derived from spectrophotometry using reagents from Trinity and PS.
Venous blood (total 3 mL) was collected from purposively selected participants with known G6PD activity in the Chittagong Hill Tracts, Bangladesh to cover the broadest possible range of G6PD activities. Samples were stored at 4 °C and G6PD activity measured within 24 h by spectrophotometry in a reference laboratory in Dhaka.
Haemoglobin (Hb) was measured by a complete blood count (CBC) using a XN-1000 (Sysmex Corporation, Kobe, Japan) immediately before spectrophotometry; samples were tested no more than an hour apart on a Shimadzu UV-1800 (Shimadzu, Japan) with kits from Trinity (Cat. No.: 345-B) and PS (Cat. No.: G7583) according to manufacturer instructions. Each kit was run in duplicate, with a measuring temperature of 30 °C for Trinity and 37 °C for PS, all measurements were done by the same two experienced and well-trained laboratory technicians who had a Master degree in a relevant field. G6PD activity was calculated from the change in absorbance at 340 nm over a period of 5 min, as per the manufacturer’s instructions, and the derived G6PD activity (U/dL) was normalized by Hb (U/gHb). G6PD deficient (Cat. No.: HC-108DE), intermediate (Cat. No.: HC-108IN) and normal controls (Cat. No.: HCS-108) (all from ACS Inc., Fishers, USA) were run daily prior to sample testing.
Repeatability was assessed per assay by correlating the first and second measurement of each kit. For the inter-assay comparison, the mean of both normalized results was calculated and compared by Students t-test, Bland Altman plots and the Pearson correlation coefficient (r).
100% G6PD activity was defined for each assay by calculating the median activity of all participants that had been found to have G6PD activity above 70% of the adjusted male median (AMM) in earlier studies [4, 8]. Each individual was then categorized as having below 10%, below 30% and below 70% activity. The proportions within each category were compared between kits using the McNemars test for correlated proportions. The sensitivity and specificity of PS were calculated at each threshold assuming Trinity as the gold standard assay .
A total of 50 samples were analysed by both kits and no results were excluded from the analysis. The repeatability of Trinity (r = 0.9841, p < 0.001) and PS (r = 0.9833, p < 0.001) was very high and did not differ significantly among the assays (p = 0.904).
The results from PS correlated closely with results from Trinity (r = 0.9799, p < 0.001; Fig. 1). The mean of the activities derived by Trinity was 7.6 U/gHb (95% confidence interval (95% CI) 6.53–8.64) compared to 7.7 U/gHb (95% CI 6.67–8.75) derived by PS; p = 0.126), with 95% limits of agreement ranging from − 1.32 to 1.57 U/gHb.
100% G6PD activity was based on 24 samples and was 10.1 U/gHb (interquartile range (IQR): 8.98–11.24) for PS and 9.6 U/gHb (IQR: 8.88–11.04) for Trinity. When results were categorized, 1 discrepant result (2%) was observed (Table 1 and Fig. 1), and the proportion of samples classified as being below 30% and 70% did not differ significantly (p = 1.000 and p = 0.317 respectively). At the 30% threshold the sensitivity of PS was 100% (95% CI 59.0–100.0) with a specificity of 100% (95% CI 91.8 to 100.0), at the 70% threshold the corresponding values were 100% (95% CI 79.4–100.0) and 97.1% (95% CI 84.7–99.9) respectively.
These findings demonstrate that PS is a reliable alternative to the assays from Trinity for the quantification of G6PD enzyme activity by spectrophotometry. Intra and inter assay correlation for the assays was excellent, suggesting that a single measurement would be sufficient providing good quality control and well-informed technicians. No sample with G6PD activity below the clinically relevant 30% cut off activity was misclassified, one sample was considered below 70% by PS but not by Trinity , however the respective sample was borderline (Fig. 1). Both assays showed a very small mean difference, suggesting that population specific cut offs  established earlier by Trinity kits are likely to be applicable to results obtained using the PS kits, although further confirmatory studies are warranted.
The operational characteristics of both assays are similar, with the exception that PS requires a higher temperature if no temperature correction factor is to be applied. This minor modification can be achieved on most if not all spectrophotometers suitable for Trinity kits. The cost of the Trinity kit in 2016 was 3.60 USD/test, whereas the cost of each PS kit is currently approximately 2.00 USD/test (these prices refer to the Australian market and are likely to differ depending on location and distributor).
These results demonstrate that both assays work well if performed by well trained technicians and when maintaining quality control measures. Given that assays from Trinity are no longer available, assays from PS provide a good alternative at lower costs.
All samples were tested on a high—end machine, assays may perform at lower accuracy on different machines. Inter-reader variability was not assessed in the course of this study, however given the consistent results, good performance observed and extensive experiences of both laboratory technicians we do not believe this has impacted on our findings.
- NADP+ :
nicotinamide adenine dinucleotide phosphate
nicotinamide adenine dinucleotide phosphate hydrogen
Trinity Biotech (Ireland)
Pointe Scientific (USA)
complete blood count
United States Dollar
von Seidlein L, Auburn S, Espino F, Shanks D, Cheng Q, McCarthy J, Baird K, Moyes C, Howes R, Menard D, et al. Review of key knowledge gaps in glucose-6-phosphate dehydrogenase deficiency detection with regard to the safe clinical deployment of 8-aminoquinoline treatment regimens: a workshop report. Malar J. 2013;12:112.
Cappellini MD, Fiorelli G. Glucose-6-phosphate dehydrogenase deficiency. Lancet. 2008;371:64–74.
Maisels MJ. Managing the jaundiced newborn: a persistent challenge. CMAJ. 2015;187:335–43.
Domingo GJ, Satyagraha AW, Anvikar A, Baird K, Bancone G, Bansil P, Carter N, Cheng Q, Culpepper J, Eziefula C, et al. G6PD testing in support of treatment and elimination of malaria: recommendations for evaluation of G6PD tests. Malar J. 2013;12:391.
Ley B, Bancone G, von Seidlein L, Thriemer K, Richards JS, Domingo GJ, Price RN. Methods for the field evaluation of quantitative G6PD diagnostics: a review. Malar J. 2017;16:361.
Standardization of procedures for the study of glucose-6-phosphate dehydrogenase. Report of a WHO Scientific Group. World Health Organ Tech Rep Ser 1967, 366:1–53.
Minucci A, Giardina B, Zuppi C, Capoluongo E. Glucose-6-phosphate dehydrogenase laboratory assay: how, when, and why? IUBMB Life. 2009;61:27–34.
Ley B, Alam MS, O’Donnell JJ, Hossain MS, Kibria MG, Jahan N, Khan WA, Thriemer K, Chatfield MD, Price RN, Richards JS. A comparison of three quantitative methods to estimate G6PD activity in the Chittagong Hill tracts, Bangladesh. PLoS ONE. 2017;12:e0169930.
Watson J, Taylor WRJ, Bancone G, Chu CS, Jittamala P, White NJ. Implications of current therapeutic restrictions for primaquine and tafenoquine in the radical cure of vivax malaria. PLoS Negl Trop Dis. 2018;12:e0006440.
MSA implemented the study and coordinated patient enrolment and analysis of samples, MGK and NJ analyzed the samples and entered all collected data, RP and BL developed the study protocol, BL analyzed data and wrote the first draft of the manuscript. All authors read and approved the final manuscript.
We thank all study participants and study staff involved in this study. icddr,b is also grateful to the Governments of Bangladesh, Canada, Sweden and the UK for providing core/unrestricted support.
The authors declare that they have no competing interests.
Availability of data and materials
All relevant data are attached (Additional file 1).
Consent for publication
Ethics approval and consent to participate
All participants provided written informed consent to participate prior to enrolment. Study procedures were reviewed and approved by the ethical review committee (ERC) and research review committee (RRC) of the icddr,b (PR-17043; Bangladesh) and the Australian Human Research Ethics Committee (HREC) of the Northern Territory (HREC 17.2771; Australia).
RNP is funded by the Wellcome Trust (Senior Fellowship in Clinical Science, 200909), BL is funded by the Australian Department of Foreign Affairs and Trade and is also funded by the Bill & Melinda Gates Foundation (OPP1054404). The funders had no role in study design, data collection and analysis, decision to publish, or preparation of the manuscript.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
Rights and permissions
Open Access This article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.
About this article
Cite this article
Alam, M.S., Kibria, M.G., Jahan, N. et al. Spectrophotometry assays to determine G6PD activity from Trinity Biotech and Pointe Scientific G6PD show good correlation. BMC Res Notes 11, 855 (2018). https://doi.org/10.1186/s13104-018-3964-7