- Research article
- Open Access
A pilot study to determine the feasibility of collecting amniotic fluid samples from women during labour and measuring amniotic fluid lactate at point of care
© Hall et al.; licensee BioMed Central Ltd. 2013
- Received: 17 August 2012
- Accepted: 18 March 2013
- Published: 26 March 2013
The level of lactate in amniotic fluid may provide useful clinical information when assessing progress of a woman’s labour and if so, a rapid, reliable method to assess amniotic fluid lactate is required in order to be clinically relevant. However, measuring lactate levels in amniotic fluid, using portable, handheld lactate meters may be less accurate than reference laboratory instruments designed to measure lactate levels in aqueous solutions. Prior to conducting a large study, we assessed recruitment, consent and sampling procedures, and the accuracy of a handheld lactate meter to measure lactate in amniotic fluid. We compared amniotic fluid lactate results obtained using the hand held Lactate Pro (Arkray) to results obtained using reference laboratory methods ABX Pentra 400 (Horiba).
We recruited 35 nulliparous women during their antenatal hospital visits and tested amniotic fluid samples collected from 20 labouring women. The handheld Lactate Pro meter was found accurate from 9–20 mmol/L with a Passing & Bablok regression of y = 0.18 + 0.97x (95% CI 0.76–1.45). Amniotic fluid lactate results remained reliable in the presence of potential contaminants commonly encountered during labour; obstetric lubricant, blood and meconium.
The measurement of amniotic fluid lactate using the Lactate Pro meter was reliable compared to reference laboratory methods for measuring lactate levels in amniotic fluid. The pilot study enabled the refinement of information, recruitment, consenting and sampling procedures prior to commencing a large cohort study.
- Amniotic fluid
- Point of care
Lack of progress in labour (dystocia) is one of the most commonly occurring problems requiring intervention during labour. In many industrialized countries labours are treated for dystocia by augmenting with oxytocin to accelerate or improve progress[2, 3]. However there is currently no accurate method for predicting at the onset of augmentation who will progress to a vaginal birth or who will require a cesarean section. In clinical practice, many labours deemed lacking in progress result in a normal vaginal birth. Conversely, women augmented for slow progress in labour may experience long painful labours that ultimately result in caesarean section.
Currently dystocia is diagnosed clinically by observing factors such as slowing of cervical dilation, slowing of descent and rotation of the fetal head, and changes in strength, duration and frequency of uterine contractions. In the absence of a clear and consistent definition or marker to guide decision making there is considerable variation in diagnosis and clinical practice. There is a need for improved diagnostic methods and decision making tools in the diagnosis of dystocia in order to reduce the high rates of labour intervention attributable to this condition.
Measurement of lactate in amniotic fluid of labouring women may be a surrogate biochemical marker for labour dystocia[6–9] based on the reasonable hypothesis that dystocia is partly due to lactic acidosis of myometrial tissue. Recent physiological studies on the effects of acidification of the uterus show human myometrium to be sensitive to changes in pH, and that accumulation of lactic acid in myometrium during contractions may reduce the strength of contractions[7, 11]. More recent studies of the acid–base balance of uterine tissue indicate that MCT 4 proteins, which appear to be activated when the myometrium is hypoxic, may be the mechanism by which lactic acid in myometrium is transported to the surrounding amniotic fluid.
If amniotic fluid lactate is relevant in the clinical management of dystocia, it is essential that a rapid and accurate test for amniotic fluid lactate is available. Although measurement of lactate using laboratory based spectrophotometric and fluorometric assays is possible this is not clinically useful in diagnosing dystocia due to time constraints in the clinical setting of active labour. The Lactate Pro is a small, portable hand held device used to process blood samples at the point of care which is relatively inexpensive. It has the potential to be used in rural and remote settings where laboratory technology may be unavailable due to costs and /or operator skill and has been found to be easy to operate and accurate[13, 14]. Manufacturer’s specifications indicate the Lactate Pro meter measures lactate concentrations in whole blood between 0.8 and 23 mmol/L and previous tests of reliability in amniotic fluid show minimal measurement error with a coefficient of variation of (1.7–3.0%). However, validation studies for use of the Lactate Pro to test lactate in amniotic fluid are not well described, and previous studies of amniotic fluid lactate not sufficiently powered to determine the utility of amniotic fluid lactate measurement for women in labour.
This report describes an assessment of a handheld meter (the Lactate Pro) for measuring amniotic fluid lactate, prior to conducting a large prospective study. The primary objectives of the pilot amniotic fluid lactate study were to assess the consenting processes for recruiting women to amniotic fluid lactate research and evaluate methods of amniotic fluid collection. We also aimed to determine the clinical aspects of relevant confounding factors in the measurement of lactate in amniotic fluid by contaminants, and determine the concordance between bedside measurement of lactate using the Lactate pro hand held meter and reference laboratory biochemical measurements.
Pilot study population and eligibility criteria
Women who booked to give birth at a tertiary teaching hospital between October 2011 and January 2012 were considered for inclusion. Nulliparous women with a live, singleton cephalic fetus and a spontaneous onset or induction of labour at 37–42 weeks gestation were eligible to participate. A final decision regarding inclusion was made when the woman was in labour to ensure the fetus remained in a cephalic presentation and was not compromised, as evidenced by routine admission cardiotocography. Women were excluded if they had a known fetal anomaly or a recognised contraindication to vaginal birth at the onset of labour. Three women with a singleton fetus with polyhydramnios undergoing elective amniocentesis in the third trimester of pregnancy were also recruited, in order to collect amniotic fluid under sterile conditions. The Study Protocol was approved by New South Wales Northern Network Human Research Ethics Committee 11/140 through the National Ethics Application Form Submission AU/19E1A08.
Recruitment, consent and enrolment
Recruitment took place at antenatal visits at or after 36 weeks gestation at the Outpatient Department of a public tertiary level teaching hospital. When a woman who had consented to participate in the study arrived at hospital in labour, or for an induction, the research midwife was notified by staff. The research midwife confirmed that all inclusion criteria and none of the exclusion criteria were present and the woman was enrolled into the study, after giving verbal confirmation of her consent to participate.
Sample collection and amniotic fluid testing in the birthing room
Amniotic fluid samples were collected from labouring women at the following defined time points after arrival at the hospital. On admission if membranes were already ruptured; at spontaneous rupture of membranes during labour; at the time of artificial rupture of membranes; at routine vaginal assessment of labour progress (usually four hourly); prior to augmentation of labour; and at the time of birth. Amniotic fluid was not collected if a woman had not ruptured her membranes.
Non-invasive methods for collecting amniotic fluid were trialled by the research midwife. It became apparent that amniotic fluid was most easily collected at the time of vaginal examination as the fluid pooled in the cupped, gloved, palm of the examining clinician. Results were obtained either by touching the primed Lactate Pro test strip directly to the pooling fluid during the examination or, by collecting some of the pooling fluid in a needle-less syringe. Alternatively, where women had copious flowing amniotic fluid, lactate was measured by applying the primed test strip directly to a freshly applied perineal pad. In collections where a syringe was used, if sufficient fluid was available, after testing with the Lactate Pro, a portion of the sample was sent to the laboratory for reference testing. In contrast, studies by Wiberg Itzel et al. (2008) obtained amniotic fluid by inserting a silicon catheter into the labouring uterus, via the vagina. We did not adopt or test this method of obtaining amniotic fluid as we aim to determine if amniotic fluid lactate, obtained at the time of routine examination by non-invasive methods, is a potential biomarker for clinicians to assess progress of labour.
Amniotic fluid specimens were tested at the time of collection using the Lactate Pro analyser according to manufacturer’s instructions. Briefly, a test strip was inserted into the Lactate Pro, ensuring correct orientation. The device, with test strip in situ was dipped in a vertical position into the amniotic fluid sample. Results, available within 1 minute were recorded by the research midwife. In most cases, clinical staff remained blinded to the result, except for teaching purposes where the research midwife shared the results with clinical staff. Amniotic fluid lactate results were never considered in clinical decision making.
Validation of point of care amniotic fluid lactate measurement
Amniotic fluid lactate results were compared to validated reference laboratory techniques for measuring lactate in aqueous fluids, the ABX Pentra 400 (Horiba Medical) using the following methods.
At the time of point of care testing with the Lactate Pro, where sufficient amniotic fluid sample was available, amniotic fluid was collected via syringe and inserted into sodium fluoride preservative tubes (Becton Dickinson). Within one hour of collection amniotic fluid samples were centrifuged at 8000×g for 2 minutes. The supernatant was removed and stored at −20°C until analysis by the ABX Pentra 400, courtesy of the Clinical Chemistry Laboratory at South Eastern Area Laboratory Services, Randwick.
Triplicate saline samples spiked for predicted Lo, Med and Hi results
Lactate 62 mmol/L
1.5 ± 0.2
7.0 ± 0.1
21.1 ± 0.2
Saline + fluoride preservative
1.6 ± 0.6
6.8 ± 0.1
19.8 ± 0.2
A comparison between Lactate Pro meter results and the automated lactate analyser (Abx Pentra 400) of amniotic fluid collected under sterile conditions and spiked with a known quantity of lactate
[Lactate] (mmol/L) Lactate Pro
[Lactate] (mmol/L) AbX Pentra 400
9.5 ± 0.1
10.0 ± 0.2
10.0 ± 0.1
10.6 ± 0.5
10.5 ± 0.5
10.5 ± 0.1
14.6 ± 0.6
14.2 ± 0.3
21 ± 0.8
18.4 ± 0.1
26.3 ± 0.7
Assessment of variations in lactate concentration using the Lactate Pro in the presence of common obstetric contaminants
Additive to amniotic fluid#
Concentration of additive
[Lactate] (mmol/L) Lactate Pro
6.3 ± 0.9
5.9 ± 0.3
6.1 ± 0.3
6.4 ± 0.4
7.9 ± 0.5
6.3 ± 0.9
6.4 ± 0.5
6.0 ± 0.5
5.9 ± 0.3
5.0 ± 0.1
7.8 ± 0.0
7.7 ± 0.3
7.4 ± 0.2
7.5 ± 0.3
7.7 ± 0.3
As unintended delays may occur between the collection and testing of samples in the clinical environment, we performed a time series to determine the effect of time delays (on lactate concentration). When sufficient quantity of sample was available, after ascertaining baseline amniotic fluid lactate at point of care, a drop of the sample remaining in the collecting syringe was re-tested after 5, 15 and 30 minutes to identify possible degradation of lactate. Lactate stability studies were extended to 24 hours after collection of two amniotic fluid samples collected under sterile conditions (one spiked with 10 mmol/L lactate and one sample unspiked); the spiked, non-centrifuged amniotic fluid samples were stored overnight at 4°C, then tested after 1, 5, 15 and 30 minutes at room temperature.
Validation of lactate Pro as an accurate measure of amniotic fluid lactate
The lactate concentrations in saline samples with and without fluoride preservative were almost identical (Table 1). While average lactate concentrations measured were similar, there was more variation in the samples with preservative. This may have been due to a problem of homogenization of the sample with the sodium fluoride crystals, most likely a preanalytical problem than an analytical variation of the device at low concentrations.
Assessment of potential confounders of lactate levels in amniotic fluid
Amniotic fluid lactate increased in the presence of whole blood (which has its own lactate load) and decreased in the presence of other contaminants as the ratio of contaminant to amniotic fluid increased (Table 3). In the time series analysis, samples collected during labour and under surgical conditions indicated that lactate concentrations remained stable with no significant reduction in measured lactate up to 30 min after initial collection, and remained stable in samples refrigerated overnight and tested at least 24 hours after collection (data not shown).
In total, 35 women were recruited to the study and 90 attempts were made to collect amniotic fluid from 20 labouring women who gave birth during the study timeframe. Amniotic fluid was not able to be collected from 15 women as nine arrived in hospital in advanced labour, three had reduced liquor volume diagnosed by ultrasound, two had not given birth at the time of data analysis and one woman gave birth with intact membranes. A lactate result was measured in 65/90 attempts to collect amniotic fluid. Mucous samples appeared to prevent the capillary action of the Lactate Pro test strip and no results were obtained from mucousy samples. Nor were results obtained at the time of artificial rupture of membranes performed to induce labour on 10 occasions; despite the procedure, sufficient amniotic fluid for sampling was not detectable at that time. Several other factors impacted on successful readings for example, one ‘Lo’ result was obtained from a sample collected directly after a woman exited the bath (presumably bathwater contamination). Two ‘Lo’ results were attributable to operator error (failure to prime the test strip by inserting it into the meter prior to collecting the sample). Outlier ‘Lo’ results were removed from the analysis.
Results show the handheld Lactate Pro meter to be reliable measuring lactate in amniotic fluid samples collected and tested in controlled conditions. A Passing & Bablok regression line applied to the lactate measures shows the Lactate Pro meter accurate from 9–20 mmol/ when measuring lactate in amniotic fluid collected under sterile conditions, compared to results from the reference analyser. A Passing Bablok regression was chosen due to the small sample size, as it allows measurement error (imprecision) in both the X and Y variables, does not assume measurement error is normally distributed, and is robust against outliers. The result compares favourably with correlation reported by Tanner et al. comparing the accuracy of the Lactate Pro against a laboratory-based a Radiometer ABL 700 analyser.
Our results support previous findings by others who have assessed the Lactate Pro in amniotic fluid lactate research[8, 9, 15] although we clearly outline the potential effects of contaminants on lactate levels and describe collection methods that reflect clinical practice, not previously described to our knowledge. Reassuringly, we established that possible contaminating substances encountered during labour do not alter amniotic fluid lactate results to any great extent and our small time series leads us to conclude that delays encountered in the clinical situation, prior to testing, do not unduly affect results.
Although the Amniotic Fluid Lactate™ monitoring system is now available on the market prior to conducting clinical trials to demonstrate the efficacy of amniotic fluid lactate in obstetric clinical decision making, we remain interested in exploring the utility of handheld meters to test amniotic fluid lactate for the reasons of cost efficiency, potential use in remote settings and versatility, as handheld lactate members have also been clinically evaluated to test fetal scalp and umbilical cord lactate[18, 19].
This small pilot study establishes proof of principle that the Lactate Pro meter, a commercially available product validated to measure lactate in whole blood, is a reliable measure of lactate in amniotic fluid, in samples collected from women during labour after spontaneous or artificial rupture of membranes. Moreover, we found women interested and willing to participate in amniotic fluid lactate research, although it will be necessary to reduce selection and sampling bias in future amniotic fluid lactate research studies, in order to achieve valid results about the association of lactate results with labour dystocia.
Dr Rita Horvath (Clinical Director) and Mr Anthony Diamond (Laboratory Manager) of The Clinical Chemistry Laboratory, South Eastern Area Laboratory Services, Randwick, are thanked for their professional advice and technical assistance.
Funding was received from the University of Sydney Research Support Grant, 2012.
- Lowe N: A review of factors associated with dystocia and caserean section in nulliparous women. Journal of Midwifery Womens Health. 2007, 52 (3): 216-228. 10.1016/j.jmwh.2007.03.003.View ArticleGoogle Scholar
- Li Z, McNally L, Hilder L, Sullivan E: Perinatal statistics series. Australia’s Mothers and babies. 2009, Sydney: Australian Institute of Health and WerlfareGoogle Scholar
- Kjaergaard H, Olsen J, Ottesen B, Nyberg P, Dykes A-K: Obstetric risk indicators for labour dystocia in nulliparous women: a multi-centre cohort study. BMC Pregnancy Childbirth. 2008, 8 (1): 45-10.1186/1471-2393-8-45.PubMedPubMed CentralView ArticleGoogle Scholar
- National Institute for Health and Clinical Excellence: NICE clinical guidelines. vol. 55. Intrapartum care: care of healthy women and their babies during childbirth. 2007, National Institute for Health and Clinical Excellence, Accessed March 25, 2012http://publications.nice.org.uk/intrapartum-care-cg55/guidance,Google Scholar
- Neal JL, Lowe NK, Patrick TE, Cabbage LA, Corwin EJ: What is the slowest-Yet-normal cervical dilation rate among nulliparous women with spontaneous labor onset. J Obstet Gynecol Neonatal Nurs. 2010, 39: 361-369. 10.1111/j.1552-6909.2010.01154.x.PubMedPubMed CentralView ArticleGoogle Scholar
- Akerud H, Ronquist G, Wiberg-Itzel E: Lactate distributionin culture medium of human myometrial biopsies incubated under different conditions. Am J Physiol Endocrinol Metab. 2009, 297 (E): 1414-1419.View ArticleGoogle Scholar
- Wray S: Insights into the uterus. Exp Physiol. 2007, 92 (4): 621-631. 10.1113/expphysiol.2007.038125.PubMedView ArticleGoogle Scholar
- Wiberg-Itzel E, Pettersson H, Andolf E, Hansson A, Winbladh B, Akerud H: Lactate concentration in amniotic fluid: a good predictor of labor outcome. Eur J Obstet Gynecol Reprod Biol. 2010, 152: 34-38. 10.1016/j.ejogrb.2010.05.005.PubMedView ArticleGoogle Scholar
- Wiberg-Itzel E, Pettersson H, Cnattingius S, Nordstrom L: Association between lactate concentration in amniotic fluid and dysfunctional labor. Acta Obstetricia et Gynecologica Scand. 2008, 87 (9): 924-928. 10.1080/00016340802295636.View ArticleGoogle Scholar
- Quenby S, Pierce S, Brigham S, Wray S: Dysfunctional labour and myometrial lactic acidosis. Obstet Gynaecol. 2004, 104 (4): 718-723.View ArticleGoogle Scholar
- Taggart M, Wray S: Hypoxia and smooth muscle function: key regulatory events during metabolic stress. J Physiol. 1998, 509: 313-325.View ArticleGoogle Scholar
- Ullah M, Davies A, Halestrap A: The plasma membrane lactate transporter MCT 4, but Not MCT1, is up regulated by hypoxi through a HIF-1_ dependent mechanism. Joural of Biology and Chemistry. 2006, 281 (14): 9030-9037.View ArticleGoogle Scholar
- Pyne D, Boston T, Martin D, Logan A: Evaluation of the Lactate Pro blood lactate analyser. European Journal Of Applied Physiology. 2000, 82 (1–2): 112-116.PubMedView ArticleGoogle Scholar
- Tanner R, Fuller K, Ross M: Evaluation of three portable blood lactate analysers: lactate Pro, lactate scout and lactate plus. European Journal Of Applied Physiology. 2010, 109 (3): 551-559. 10.1007/s00421-010-1379-9.PubMedView ArticleGoogle Scholar
- Wiberg-Itzel E, Cnattingius S, Nordstrom L: Lactate determination in vaginal fluids: a new method in the diagnosis of prelabour rupture of membranes. BJOG. 2005, 112 (6): 754-758. 10.1111/j.1471-0528.2004.00521.x.PubMedView ArticleGoogle Scholar
- Wiberg-Itzel E, Lipponer C, Norman M, Herbst A, Prebensen D, Hansson A, Bryngelsson AL, Christoffersson M, Sennstrom M, Wennerholm UB: Determination of pH or lactate in fetal scalp blood in management of intrapartum fetal distress: randomised controlled multicentre trial. BMJ. 2008, 336 (7656): 1284-1287. 10.1136/bmj.39553.406991.25.PubMedPubMed CentralView ArticleGoogle Scholar
- ObsteCare: The solution for dysfunctional labour. Accessed March 25, 2013http://www.obstecare.com/index.html,
- Nordstrom L: Lactate measurements in scalp and cord arterial blood. Current Opinion in Obstetrics and Gynaecology. 2001, 13: 141-145. 10.1097/00001703-200104000-00008.View ArticleGoogle Scholar
- Pennell CE, Tracy MB, Donald MC: Accurate lactate-a new method for rapid pont of sample measurement of lactate in fetal and neonatal blood. Neonatal Intensive Care. 2000, 13 (4): 53-58.Google Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.